Detailed Product Description
(1) Used Bruker D8 Advance Eco for multipurpose solutions diffractometer. Full-sized goniometer class powder XRD under ambient and non-ambient conditions.
Bruker D8 Advance Eco
General Diffraction System
Tab. 1: Technical Specifications: General Diffraction System.
Weight of complete system ready for operation (includes: generator, cooling unit, electronics, goniometer with standard sample stage, X-ray source, X-ray optics, scintillation detector) 770 kg
Weight of the enclosure and cabinet without equipment (goniometer, sample stage, X-ray source, X-ray optics, detector) 620 kg
Area load per m2 of the floor (ready for operation, system standing on integrated casters: 66cm x 90cm) 1300 kg/m²
Clearance of doors for transportation 86 cm (34″)
Height (Cabinet and Enclosure ready for operation)1868 mm (73,5″)
Width (Cabinet and Enclosure ready for operation)1300 mm (51,2″)
Depth (Cabinet and Enclosure ready for operation)1135 mm (44,7″)
Space inside enclosure: Height 956 mm (37.6″)
Space inside enclosure: Width 1014 mm (39.9″)
Space inside enclosure: Depth 1037 mm (40.8″)
Transport size on casters: Height 1300 mm (51.2″)
Transport size on casters: Width 1868 mm (73.5″)
Transport size on casters: Depth 854 mm (33.6″)
Temperature operation range 15 – 35°C, recommended temperature range 20°C - 28°C
Maximum temperature gradient 1.0°C per hour
Relative humidity 20%-80%, condensation not allowed
Atmospheric pressure Diffraction system applicable at all terrestrial locations under atmospheric air pressure conditions
Cooling Water Requirements Minimum Flow Rate 3.6 l/min
Pressure of Cooling Water Supply 4 to 7.5 bar
Pressure of Cooling Water Discharge Pressure-free
Operating Water Temperature 10 to 25° C, 15 to 20° C recommended (avoid condensation)
Diameter of Cooling Water Supply 1/2“
Water filter (water supply) Mesh size 0.15 mm (100 mesh)
Assuming a relative humidity of 80% and a room temperature of 30 °C the dew point is 26 °C, at a room temperature of 25 °C the dew point is 21 °C, respectively. In case the head of the tube housing has a temperature equal or lower than the dew point condensation will occur.
Voltage Requirements: 208V, 60Hz, 1Ph
- Area detector (GADDS)
- Goebel mirror’s (cross coupled)
- Huber ¼ cradle for precise positioning of the specimen surface (in 3-dimensions)
- A scintillation detector
- A wide variety of beam collimators (from 0.8 mm down to 0.05 mm)
-This system is a research-grade instrument that can be used for many different types of studies.
-This system is equipped with two different tube housings.
-One utilizes a chromium x-ray tube and this configuration is normally used for residual stress and texture determinations.
-The second tube housing is configured with a copper x-ray tube and Goebel mirrors, and this configuration is normally used for structure determination and compositional studies.
(on consignment for customer)